Following my polarimetry experiments at Allen Telescope Array, on October 31 I did a polarimetric observation of the quasar 3C286 with two dishes from the array to use as a test-bed for polarimetric calibration. 3C286 is a bright, compact, polarized source, with a fractional polarization intensity of around 10% and a polarization angle of 33º over a wide range of frequencies, so it makes an ideal source for polarization calibration. It is the primary polarization calibrator for VLA. The observation duration was slightly more than 2 hours, and it was done around the transit of the source, so the parallactic angle coverage is large (around 90º).
My initial idea was to use this observation to perform a “single dish” polarization calibration of each of the dishes by separate (since the math is somewhat simpler) and then perform an interferometric polarization calibration. However, after initial examination of the data, the SNR doesn’t seem large enough to do a “single dish” calibration. The polarized signal from 3C286 is rather weak and is swamped by noise from other sources in the field and from the receiver, and also by gain variations in the receive chain.
On the contrary, the interferometric calibration has worked well, since correlating the signals from the antennas allows us to discard the uncorrelated receiver noise and to phase on the target and discard other signals from the field, by means of Earth rotation aperture synthesis.
In this post I give my analysis and results of the observation. I have done an ad hoc calibration in Python to determine the polarization leakage and measure the polarization degree and angle of the source, and also a full polarimetric calibration in CASA to compare my calibration with one obtained with professional software.
The data used in this post has been published in Zenodo as the dataset “Allen Telescope Array polarimetric observation of 3C286“.
Continue reading “Polarimetric observation of 3C286 with Allen Telescope Array”
This post belongs to a series about the activities of the GNU Radio community at Allen Telescope Array. For more information about these activities, see my first post.
The feeds in the ATA dishes are dual polarization linear feeds, giving two orthogonal linear polarizations that are called X and Y and (corresponding to the horizontal and vertical polarizations). In the setup we currently have, the two RF signals from a single dish are downconverted to an IF around 512 MHz using common LOs and then sampled by the two channels of a USRP N32x. Since we have two USRPs, we are able to receive dual polarization signals from two dishes simultaneously.
The two USRPs are synchronized with the 10MHz and PPS signals from the observatory, but even in these conditions there will be random phase offsets between the different channels. These offsets are caused by fractional-N PLL states and other factors, and change with every device reset. To solve this problem, it is possible to distribute the LO from the first channel of a USRP N321 into its second channel and both channels of a second USRP N320. In fact, it is possible to daisy chain several USRPs to achieve a massive MIMO configuration. By sharing the LO between all the channels, we achieve repeatable phase offsets in every run.
During the first weekends of experiments at ATA we didn’t use LO sharing, and we finally set it up and tested it last weekend. After verifying that phase offsets were in fact repeatable between all the channels, I did some polarimetric observations of GNSS satellites to calibrate the phase offsets. The results are summarised in this post. The data has been published in Zenodo as “Allen Telescope Array polarimetric observation of GNSS satellites“.
Continue reading “ATA polarimetry test with GNSS satellites”
Since the beginning of October, together with a group of people from the GNU Radio community, we are doing some experiments and tests remotely at Allen Telescope Array (ATA). This amazing opportunity forms part of the recent collaboration agreement between SETI Institute and GNU Radio. We are taking advantage of the fact that the ATA hardware is relatively unused on weekends, and putting it to good use for our experiments. One of the goal of these activities is to put in contact GNU Radio people and radio astronomy people, to learn from each other and discover what features of GNU Radio could benefit radio astronomy and SETI, particularly at the ATA.
I’m very grateful to Wael Farah, Alex Pollak, Steve Croft and Ellie White from ATA and SETI Institute for their support of this project and the very interesting conversations we’ve had, to Derek Kozel, who is Principal Investigator for GNU Radio at SETI, for organizing and supporting all this, and to the rest of my GNU Radio teammates for what’s being an excellent collaboration of ideas and sharing of resources.
From the work I’ve been doing at ATA, I already have several recordings and data, and also some studies and material that I’ll be publishing in the near future. Hopefully this post will be the first in a series of many.
Here I will speak about one of the first experiments I did at ATA, which is a recording of one Galileo GNSS satellite using two of the dishes from the array. This kind of recording can be used to perform interferometry. GNSS satellites are good test targets because they have strong wideband signals and their location is known precisely. The IQ recording described in this post is published as the dataset “Allen Telescope Array Galileo E31 RF recording with 2 antennas and 2 polarizations” in Zenodo.
Continue reading “GNSS interferometry at Allen Telescope Array”